The Elcometer range of digital coating thickness gauges has been specifically designed to provide highly accurate, reliable and repeatable coating thickness. The Elcometer Coating Thickness. Gauge is an incredibly versatile gauge. With a range of probes in both Integral or Separate probe versions for coating. Elcometer Coating Thickness Gauge. Operating Instructions. English. R. English Equipment described in these instructions is covered by the following.
|Published (Last):||21 March 2015|
|PDF File Size:||8.70 Mb|
|ePub File Size:||11.78 Mb|
|Price:||Free* [*Free Regsitration Required]|
This action also serves to switch the statistics feature on.
Do not remove the probe while the instrument is switched on. Seller assumes all responsibility for this listing.
Probe life depends on the number of measurements taken and the manner in which readings are taken. Clear the error message and repeat the adjustment procedure.
With a range of probes in both Integral or Separate probe Back to home page. This is useful when there is significant variation from one reading to the next, such as lecometer occur on a rough surface or profile. Manual Holliday Detector Elcometer Documents. Line up the 2 red marks when inserting the plug.
Elcometer 345 Digital Coating Thickness Gauge
To re-adjust the gauge, repeat all the steps in Section 5. Select a valid country. Separate probes can be fitted by the user. Hold the gauge with the probe pressed firmlyb against the surface you are measuring. The eddy current principle is used for non-conductive coatings on non-ferrous metals substrates. If you do stumble upon a link that isnt functioning, do tell us about that and we will try to answer you as fast as Our products cover all aspects of coating inspection, from development through application to elcimeter application inspection.
To clear the statistical data when starting a new set of data press display will show CLs to indicate Clear Statistics?.
Elcometer Digital Coating Thickness Gauge | eBay
An intermediate foil thickness can also be used to confirm the performance over the calibrated range. If the probe is left on the bare substrate or moved quickly to the calibration foil the display will show elclmeter sequence of three rising – symbols and sound a series of bleeps as a warning. Read more about the condition.
No part of this Document may be reproduced, transmitted, transcribed, stored in a retrieval system or otherwise or translated into any language, in any form or by any means electronic, mechanical, magnetic, optical, manual or otherwise without the prior written permission of Elcometer Limited.
Press to exit menu or exit automatically when all 5 items have been entered.
The symbol will flash on the display. Consult your local Elcometer supplier. Learn more – opens in a new window or tab. To exit the list press. Contact the seller – opens in a new window or tab and request a postage method to your location.
See all condition definitions – opens in a new window or tab The Elcometer Coating Thickness Gauge is powered by internal dry cell batteries. Elcometer User Manual for Mac comes with powerful features such as editing Documents. In addition, a number of symbols can appear on the display: This item will be sent through the Global Shipping Programme and includes international tracking. To delete the last reading from the statistical calculations when the statistics feature is switched on, press and hold for more than 2 seconds but less than elcomdter seconds.
Marshall Manual – Manual Please enter a number less than or equal to 1. Repeated measurements can now be taken on this foil and the display will show the accumulated average of these readings. Previous readings can not be deleted individually. Press and release to begin the special calibration sequence. Elcometer Protective Coatings Inspection Manual For gauge checks and certification contact Elcometer or your local Elcometer supplier. Confirm that the average gauge reading on each calibration foil is correct.
For further details see EN ‘Paints and Varnishes – Corrosion protection of steel structures elometer protective paint systems – Measurement and acceptance criteria of the dry film thickness. Elcometer Manual Download Report.
Details of Elcometer offices around the world are given on the outside cover of these Operating Instructions. The Elcometer Coating Thickness Gauge is a world elcometre product. Remove the batteries from the gauge if it is to remain unused for a long period of time.
The display will blink then re-display the reading.
Elcometer S Operating Manual Documents. If the following display appearsFRIntegral probe0RSeparate probethe value is just greater than the range of the gauge, or the gauge is out of calibration. When the batteries are exhausted, the gauge will beep three times and then switch off. This is either due to a very high electrical noise environment, beyond emc specification limits, or could indicate a gauge fault.
With the gauge switched on, press. If the key is held for more than 5 seconds the calibration routine is activated. Before you use your gauge for the first time, remove the paper strip from the battery compartment.